|Published (Last):||4 January 2008|
|PDF File Size:||8.23 Mb|
|ePub File Size:||13.14 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Start Chapter 1 prepares the multimeter for use and helps you get familiar with a few of its front-panel features. Maintenance If any portion of the Test Lead Set is worn or damaged, do not use. For continued protection against fire, replace manaul line fuse and the current-protection fuse only with fuses of the specified type and rating.
No part of this manual may be reproduced in any form agilent 34401a user manual pdf download by any means including electronic storage and retrieval or translation into a foreign language without prior agreement and written consent from Agilent Agioent, Inc.
Do not use detergent or chemical solvents. Return the product to an Agilent Technologies Sales and Service Office for service and repair to agilent 34401a user manual pdf download that safety features are maintained.
This guide is valid for the firmware that was installed in the instrument at the time of manufacture. Measurement Tutorial Chapter 7 discusses measurement considerations and techniques to help you obtain the best accuracies and reduce sources of measurement error. Further, to the maximum extent agilent 34401a user manual pdf download by applicable law, Agilent disclaims all warranties, either express or implied, with regard to this manual and any information contained agilent 34401a user manual pdf download, including but not limited to the implied warranties of merchantability and fitness for a particular purpose.
To review the display annunciators, hold down the Shift key as you turn on the multimeter. To ensure safe operation of the instrument, do not exceed the Protection Limits shown on the front and rear panel, and defined as follows:.
With reference to the equipment types in the WEEE directive Annex 1, this product is classified as a “Monitoring and Control instrumentation” product.
The current-protection circuitry includes a fuse on the rear panel. The menu is organized in a top-down tree structure with three levels.
34401A Digital Multimeter, 6½ Digit
The Protection Limit is Vpk for all of the terminal pairings:. Two Protection Limits are defined for these terminals: Most keys have a shifted function printed in blue above the key.
To ensure safe operation of the instrument, do not exceed the Protection Agilent 34401a user manual pdf download shown on the front and rear panel, and defined as follows: Each listing contains enough information to help you usr and solve the problem.
Error Messages Chapter 5 lists the error messages that may appear as you are working with the multimeter. Only qualified, service-trained personnel should remove the cover from the instrument.
Agilent a Multimeter User Manuals
However, the A may not be used with its HI and LO inputs connected agilent 34401a user manual pdf download mains in permanently installed electrical devices such as the main circuit-breaker panel, sub-panel disconnect agilent 34401a user manual pdf download, or permanently wired motors. Restricted Rights Legend U. The A may be used to make measurements with the HI and LO inputs connected to mains in such devices, or to the branch outlet itself up to VAC.
Do not operate this switch while signals are present on the front or rear terminals. Also, do not use a damaged or worn Test Lead Set. This declaration covers all options of the above product s. The switch is not intended as an active multiplexer. Do not use this product in any manner not specified by the manufacturer.
The protective features of this product may be impaired if it is used in a manner not specified in the operation instructions.
Multimeter is waiting for a single trigger or external trigger. Clean the outside of the instrument with a soft, lint-free, slightly dampened cloth.
If you dlwnload questions relating to the operation of the Agilent A, call in the United States, or contact your nearest Agilent Sales Office. For example, to select the dc current function, press Shift DC V. Multimeter is agilwnt continuity test function. If your A fails within one year of purchase, Agilent will repair or replace it free of charge. Agilent shall not be liable for errors or for incidental or consequential damages in connection with the furnishing, use, or performance of this document or of any information contained herein.
To avoid the danger of electric shock, do not connect the inputs to mains for line voltages above Pd. The LO amnual terminal can safely “float” a maximum of Vpk relative to ground. However, upgrading the firmware may 43401a or change product features. Then, press the key that has the desired label above it. Agilent provides this customary commercial license in Software and technical data pursuant to Downloaad IEC Measurement Category II Overvoltage Protection To protect against the danger of electric shock, the Agilent A Digital Multimeter provides overvoltage protection for line-voltage mains connections meeting both of the following conditions: The HI and LO input terminals are used for voltage, resistance, frequency periodand diode test measurements.
Multimeter is in diode test function. The Agilent A are designed to safely withstand occasional transient overvoltages agilent 34401a user manual pdf download to Vpk. Agilent 34401a user manual pdf download is implied by the above limits, the Protection Limit for the HI input terminal is a maximum of Vpk relative to ground.
For further information, please contact your local Agilent Technologies sales office, agent or distributor. The Agilent A Digital Multimeter provides protection circuitry to prevent damage to the instrument and to protect against the danger of electric shock, provided the Protection Limits are not exceeded.
It calls mahual to an operating procedure, practice, or the like that, if not correctly performed or adhered to, could result in personal injury or death. To avoid instrument damage and the risk of electric shock, do not exceed any of the Protection Limits defined in the following section. Any interruption of the protective grounding conductor or disconnection of the protective earth terminal will cause agilent 34401a user manual pdf download potential shock agilent 34401a user manual pdf download that could result in personal injury.
Safety Information General Do not use this product in agient manner not specified by the manufacturer. User Guides and Service Manuals. Conforms with the following product standards: The Protection Limit is Vpk for all of the terminal pairings: